Add Attachments to Content Item ‘ASTM D Glass Transition Temperature ( DMA Tg) of Polymer Matrix Composites by Dynamic Mechanical Analysis (DMA) . ASTM D Glass Transition Temperature (DMA Tg) of Polymer ASTM D has become a widely used test standard in many companies due to. ASTM D用DMA测试Tg_能源/化工_工程科技_专业资料。Designation: D – 07?1 Standard Test Method for Glass Transition.
|Published (Last):||13 February 2015|
|PDF File Size:||9.48 Mb|
|ePub File Size:||20.14 Mb|
|Price:||Free* [*Free Regsitration Required]|
Strains of less than 0.
Before the transition the storage modulus is relatively constant, the sigmoidal change during transition is clear, and after the transition the storage modulus is relatively constant. If you feel that your comments have not received a fair hearing you should make your views known to the ASTM Committee on Standards, at the address shown below.
Depending on the analyzer, typical specimen size can range from 56 6 4 3 12 6 1 3 2. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use. D — 07? Measure the specimen length to 0. This could indicate that the sample was taken from a defective portion of the laminate. A non-standard specimen con?
Summary of Test Method 4. The conditioned specimens shall be tested within 30 minutes after removal from the conditioning chamber, or stored in sealed MIL-PRF or equivalent aluminized bag until test.
This standard is subject to revision at any time by the responsible technical committee and must be reviewed every? Interpretation of Results Current edition approved Dec.
It is possible that more than one transition exists, but this should be con? The equipment must be calibrated in the same loading mode as will be used for testing, either dual cantilever or three-point bending. This temperature is designated as TB. D Terminology for Plastics: The peak temperature of the tangent delta curve Tt is identi? Work Item s – proposed revisions of this standard.
Weigh the specimen to the nearest milligram 0. If another equipment type or loading mode is used the non-standard approach shall be described in the report and the test result d7082 as non-standard. The materials listed in Table 1 are de? If the testing is part of a designed experiment, other sampling techniques may be used if described in the asfm plan. The DMA Tg value is frequently used to indicate the upper use temperature of composite materials, as well as for quality control of composite materials.
The temperature calibration points must span the DMA Tg result.
ASTM D7028 – 07(2015)
Any deviations from the standard orientation shall be reported and the test results noted as non-standard. Historical Version s – view previous versions of standard. A change in heating rate will affect the glass transition temperature result; the standard heating rate is the best available compromise for comparing DMA Tg results of dry and wet laminates.
Link to Active This link will always route to the current Active version of the standard. Other loading modes such as tensile, torsion or shear may produce different test results. Test Method E speci? Typically the thermocouple should be as close to the sample as possible.
If a non-standard frequency, or the resonance mode, is used, the non-standard approach shall be described in the report and the test result recorded as non-standard.
The DMA Tg is determined to be the intersection of two tangent lines from the storage modulus by this test method.
Should the results be retained, then these conditions shall be described in the test report. Your comments will receive careful consideration at a meeting of the responsible technical committee, which you may attend. Mount the specimen in dual cantilever mode or three-point bending mode.
The temperature at which a signi? Laminate edges or other irregularities created in the laminate by mold or bagging techniques should be avoided.
In this example the transition is less aastm than the thermogram of Fig.
For example, a linear plot scale will result in a lower value of DMA Tg. Individual reprints single or multiple copies of this standard may be obtained by contacting ASTM at the above address or at phonefaxor service astm. Sampling and Test Specimens 8.
Generally, the specimen should be qstm between the clamp faces and be parallel to the base of the instrument. Before the transition the storage modulus continues to slope downward and after the transition the storage modulus continues to slope downward.
The maximum time between cure lamination and testing shall be 30 days, after which, prior to testing, specimens shall be oven-dried in accordance with d708 In general, for a given material, a higher testing frequency produces a higher DMA Tg value than this standard, while use of the resonance mode will yield a different DMA Tg that may be either higher or lower than the standard. For some types of dynamic mechanical analyzers, a purge gas?
Weigh the specimen after the test to the nearest milligram 0. Therefore, it is recommended that the test specimens should be weighed before and after DMA Tg testing to quantify the moisture change in the specimen resulting from the DMA Tg test.
The same conditions shall be used for both calibration and testing runs. During the glass transition, the storage r7028 of the composite material is signi?